Table 4.

Analysis of variance for 73 renal transplant patients measured twice on TBW (by D2O dilution and MF-BIA) supplemented with estimation of variance componentsa

Sources of VariationSSdfMSEMS = σ2ICC/Proportion
ICC = method agreement = proportion of patient VC to the total amount of patient VC plus the patient-by-method VC,
or: ICC = σ2P/(σ2P + σ2P × M)
For TBW: σ2P = (MSP - MSP × M)/2
σ2M = (MSM - MSP × M)/73
σ2P × M = MSP × M
SDD = smallest detectable difference; with two repeated measurements and a probability of 95%
SDD = √MSP × M × √2 × 1.96
For TBW: SDD = 4.04, so with a probability of 95% score differences lie between -3.3 and 4.8.
“Limits of agreement” are almost equivalent to SDD limits. Factor “patients” is considered to be a “random” factor, the factor “method” is fixed.
aSS, sums of squares; MS, mean sum of squares; df, degrees of freedom; EMS, σ2 = expected mean sum of squares = (estimated) variance component (VC); ICC, intraclass correlation coefficient.
Patients (P)5157.977271.6434.760.943
Method (M)19.00119.00(0.23)
P × M (error)152.65722.122.120.057
Total14536.881.000